J Lab Med Qual Assur 2015; 37(3): 148-152
Published online September 30, 2015
Copyright © Korean Association of External Quality Assessment Service.
Kyong-Ho Cha, Jeong- Joong Lee, Han-Na Kim, Hyojin Chae, and Yonggoo Kim
Department of Laboratory Medicine, Seoul St. Mary’s Hospital, The Catholic University of Korea College of Medicine, Seoul, Korea
Correspondence to:Yonggoo Kim
Department of Laboratory Medicine, Seoul St. Mary’s Hospital, 222 Banpo-daero, Seocho-gu, Seoul 06591, Korea
This is an Open Access article distributed under the terms of the Creative Commons Attribution Non-Commercial License (http://creativecommons.org/licenses/by-nc/3.0) which permits unrestricted non-commercial use, distribution, and reproduction in any medium, provided the original work is properly cited.
Tacrolimus is an immunosuppressive agent used to prevent post-transplantation rejection. Tacrolimus has a narrow therapeutic window and therefore, its whole blood concentration is measured for therapeutic drug monitoring. In this report, we present two cases of falsely elevated tacrolimus concentrations identified in recipients of solid organ transplants due to analytical interferences in the antibody-conjugated magnetic immunoassay (ACMIA) method used. Tacrolimus concentrations measured using ACMIA were 4- to 8-fold higher than the values obtained using liquid-chromatography-tandem mass spectrometry (LCMS/ MS) or chemiluminescent micro-particle immunoassay. The cause of this interference remains unknown, but the identification of a possible false elevation of tacrolimus is of paramount importance in clinical practice. Pre-treatment of samples by ethanol extraction or using alternative methods of tacrolimus measurement such as LC-MS/MS are necessary to obtain reliable results in the event of an analytical interference.
(J Lab Med Qual Assur 2015;37:148-152)
Keywords: Tacrolimus, Immunoassay, Interference, Antibody-conjugated magnetic immunoassay
|View Full Text||PubReader|
|Abstract||Print this Article|
|E-mail alert||Export to Citation|
|Article as PDF||Open Access|